You can resize any of the graphs by clicking and dragging a rectangle. If you hover the mouse over the graph, you will see a pop-up showing the coordinates. One of the icons in the upper right corner will let you export the graph in other formats.
This optical absorption measurement of 9,10-Diphenylanthracene were made by R.-C. A. Fuh on 06-26-1995 using a Cary 3. The absorption values were collected using a spectral bandwidth of 1.0 nm, a signal averaging time of 0.133 sec, a data interval of 0.25 nm, and a scan rate of 112.5 nm/min.
These measurements were scaled to make the molar extinction coefficient match the value of 14,000cm-1/M at 372.5nm (Berlman, 1971).
The fluorescence emission spectrum of 9,10-Diphenylanthracene dissolved in cyclohexane. The excitation wavelength was 350nm. The quantum yield of this molecule is 1. (Berlman, 1965). This spectrum was collected by on 06-26-1995 using a .
Berlman, I. B. (1965) ÃÂ¢ÃÂÃÂHandbook of Fluorescence Spectra of Aromatic Molecules,ÃÂ¢ÃÂÃÂ Academic Press, N.Y.
Berlman, I. B. (1971) ÃÂ¢ÃÂÃÂHandbook of Fluorescence Spectra of Aromatic Molecules,ÃÂ¢ÃÂÃÂ Academic Press, N.Y.
Dixon, J. M., M. Taniguchi and J. S. Lindsey (2005), "PhotochemCAD 2. A Refined Program with Accompanying Spectral Databases for Photochemical Calculations, Photochem. Photobiol., 81, 212-213.
Du, H., R.-C. A. Fuh, J. Li, L. A. Corkan and J. S. Lindsey (1998) PhotochemCAD: A computer-aided design and research tool in photochemistry. Photochem. Photobiol. 68, 141-142.
Eaton, D. F. (1988) Reference materials for fluorescence measurement. Pure Appl. Chem. 60, 1107-1114.
Hamai, S. and F. Hirayama (1983) Actinometric determination of absolute fluorescence quantum yields. J. Phys. Chem. 87, 83-89.
Meech, S. R. and D. Phillips (1983) Photophysics of some common fluorescence standards. J. Photochem. 23, 193-217.